<table>
	<tbody>
		<tr>
			<td>Features</td>
			<td>Details</td>
		</tr>
		<tr>
			<td>Max. Weighing Capacity</td>
			<td>200 g</td>
		</tr>
		<tr>
			<td>Reproducibility</td>
			<td>Starting at 1 g: &plusmn; 0.2%, Starting at 5 g: &plusmn; 0.05%</td>
		</tr>
		<tr>
			<td>Readability</td>
			<td>1 mg, 0.01%</td>
		</tr>
		<tr>
			<td>Typical Sample Quantity</td>
			<td>5 g &ndash; 15 g</td>
		</tr>
		<tr>
			<td>Display Modes for Results</td>
			<td>Moisture content in % M and % g, Dry weight in % S and g, ATRO (ratio) in % M</td>
		</tr>
		<tr>
			<td>Temperature Range and Settings</td>
			<td>40&deg;C &ndash; 200&deg;C (in increments of 1&deg;C), Standby temperature 50&deg;C &ndash; 120&deg;C</td>
		</tr>
		<tr>
			<td>Sample Heating</td>
			<td>Infrared heating using AURI heating unit, 600 W</td>
		</tr>
		<tr>
			<td>Heating Programs</td>
			<td>Standard drying, gentle drying, MA35 mode</td>
		</tr>
		<tr>
			<td>Shutoff Parameter</td>
			<td>Fully automatic, semi-automatic, manual, timer settings</td>
		</tr>
		<tr>
			<td>Forceps for Samples</td>
			<td>For easy handling of sample pans</td>
		</tr>
		<tr>
			<td>Data Interface</td>
			<td>Mini USB port; automatic printer detection, direct transfer to Microsoft&reg; Office</td>
		</tr>
		<tr>
			<td>Data Transfer</td>
			<td>SD card; import/export methods and results</td>
		</tr>
		<tr>
			<td>Housing Dimensions (W &times; D &times; H)</td>
			<td>215 &times; 400 &times; 210 mm (8.5&rdquo; &times; 15.7&rdquo; &times; 8.3&rdquo;)</td>
		</tr>
		<tr>
			<td>Weight</td>
			<td>Approx. 6.2 kg (approx. 13.6 lbs.)</td>
		</tr>
	</tbody>
</table>
