<table>
    <tbody>
        <tr>
            <td>Brand</td>
            <td>Resmed</td>
        </tr>
        <tr>
            <td>Model</td>
            <td>Astral 150</td>
        </tr>
        <tr>
            <td>Indications For Use</td>
            <td>Life Support For Patients Over 5Kg</td>
        </tr>
        <tr>
            <td>CIRCUIT</td>
            <td>Single Circuit With Leak<br />
            Single Circuit With Valve<br />
            Double Circuit</td>
        </tr>
        <tr>
            <td>THERAPY</td>
            <td>Valve Circuit Therapy - CPAP, ACV, PACV, P-SIMV, V-SIMV, PS<br />
            Leak Circuit Therapy - CPAP, ST, PAC<br />
            Manual Breath<br />
            Sigh Breath (Recruitment)<br />
            Apnoea Ventilation - Volume And Pressure<br />
            Preset Programs - 4</td>
        </tr>
        <tr>
            <td>PRESSURE RANGE AND SETTINGS</td>
            <td>Delivered Pressure (cm H2O) - 2-50 (Leak Circuit), 0-50 (Valve Circuit)<br />
            Delivered Tidal Volume (cm H2O) - Adult 100 to 2500 ml, Paediatric 50 to 300 ml<br />
            Respiratory Rate - Adult Off 2 to 50 bpm, Paediatric Off 5 to 80 bpm</td>
        </tr>
        <tr>
            <td>DATA</td>
            <td>SpO2 Monitoring - Optional<br />
            FiO2 Monitoring<br />
            7 Days Of High-Resolution Data With 25 Hz Pressure Flow - On Screen And Download<br />
            365 Days Of Summary Data - On Screen And Download</td>
        </tr>
        <tr>
            <td>ALARMS</td>
            <td>A Wide Range Of Settable Alarms Are Available Including FiO2 And SpO2 When Connected</td>
        </tr>
        <tr>
            <td>DIMENSIONS</td>
            <td>Weight - 3.2 Kg<br />
            Dimensions (L x W x H) - 285mm x 215mm x 93mm<br />
            Internal Battery - Lithium-Ion Battery, 14.4V, 6.6 Ah, 95 Wh. 8 Hrs. Backup Power<br />
            Sound Power Level - 43 +/- 3 dBA*<br />
            Sound Pressure Level - 35 +/- 3 dBA*</td>
        </tr>
        <tr>
            <td>IEC 60601-1 CLASSIFICATION</td>
            <td>Medical Electrical Equipment - General Requirements For Basic Safety And Essential Performance -<br />
            Class II (Double Insulation) Type BF Continuous Operation Suitable For Use With Oxygen</td>
        </tr>
        <tr>
            <td>Available</td>
            <td>Rental &amp; Sale</td>
        </tr>
    </tbody>
</table>