<table>
	<tbody>
		<tr>
			<td>Brand</td>
			<td>Philips</td>
		</tr>
		<tr>
			<td>Model</td>
			<td>Trilogy 100</td>
		</tr>
		<tr>
			<td>Environmental</td>
			<td>Temperature - 5 Degree Celsius - 40 Degree Celsius<br />
			Relative Humidity - 15% - 95%<br />
			Atmospheric Pressure - 110kPa - 60kPa</td>
		</tr>
		<tr>
			<td>Physical</td>
			<td>Weight - 11 Lbs, 5 Kg<br />
			Size - 4.5 x 6.88 x 9.5 in<br />
			16.69 x 38.45 x 22.52 cm</td>
		</tr>
		<tr>
			<td>Controls</td>
			<td>Circuit Types - Active With PAP, Passive<br />
			Ventilation Types - Volume Controls, Pressure Control, Bilevel<br />
			Volume Modes - AC, SIMV (w/PS), CV<br />
			Pressure Modes - CPAP, S, S/T, T, PC-SIMV (w/PS)<br />
			IPAP - 4-50 cm H2O<br />
			EPAP - 0-25 cm H2O For Active Circuits, 4-25 cm H2O For Passive Circuits<br />
			CPAP - 4-20 cm H2O For Passive Circuits<br />
			PEEP - 0-25 cm H2O For Active Circuits, 4-25 cm H2O For Passive Circuits<br />
			Pressure Support Differential - 0-30 cm H2O<br />
			Tidal Volume - 50-2000 ml<br />
			Breath Rate - 0-60 For AC Mode, 1-60 For All Other Modes<br />
			Inspiratory Time - 0.3-0.5 Secs<br />
			Rise Time - 1-6<br />
			Ramp Start Pressure - 0-25 cm H2O For Active Circuits, 4-25 cm H2O For Passive Circuits, 4-19 cm H2O In CPAP Mode<br />
			Ramp Length - Off, 5-45 Minutes<br />
			Flex - Off, 1-3<br />
			Flow Trigger Sensitivity - 1-9 L/Min<br />
			Flow Cycle - 10% - 40%<br />
			Apnea Rate - 4-60 BPM</td>
		</tr>
		<tr>
			<td>Measured Patient Parameters</td>
			<td>Tidal Volume - 0-2000 ml<br />
			Minute Ventilation - 0-99 L/Min<br />
			Leak Rate - 0-200 L/Min<br />
			Respiratory Rate - 0-80 L/Min<br />
			Peak Inspiratory Flow - 0-200 L/Min<br />
			Peak Inspiratory Pressure - 0-99 cm H2O<br />
			Mean Airway Pressure - 0-99 cm H2O<br />
			% Patient Triggered Breaths - 0-100 %<br />
			I:E Ratio - 9.9:1 to 1:9.9</td>
		</tr>
		<tr>
			<td>Alarms</td>
			<td>Circuit Disconnect - Off, 10-60 Secs<br />
			Apnea - Off, 10-60 Secs<br />
			High Tidal Volume - Off, 50-2000 ml<br />
			Low Tidal Volume - Off, 50-2000 ml<br />
			High Minute Ventilation - Off, 1-99 L/Min<br />
			Low Min Ventilation - Off, 1-99 L/Min<br />
			High Respiratory Rate - Off, 4-80 BPM<br />
			Low Respiratory Rate - Off, 4-80 BPM</td>
		</tr>
		<tr>
			<td>Available</td>
			<td>Rental &amp; Sale</td>
		</tr>
	</tbody>
</table>
