<table>
	<tbody>
		<tr>
			<td>Brand</td>
			<td>Resmed</td>
		</tr>
		<tr>
			<td>Model</td>
			<td>Astral 150</td>
		</tr>
		<tr>
			<td>Indications For Use</td>
			<td>Life Support For Patients Over 5Kg</td>
		</tr>
		<tr>
			<td>CIRCUIT</td>
			<td>Single Circuit With Leak<br />
			Single Circuit With Valve<br />
			Double Circuit</td>
		</tr>
		<tr>
			<td>THERAPY</td>
			<td>Valve Circuit Therapy - CPAP, ACV, PACV, P-SIMV, V-SIMV, PS<br />
			Leak Circuit Therapy - CPAP, ST, PAC<br />
			Manual Breath<br />
			Sigh Breath (Recruitment)<br />
			Apnoea Ventilation - Volume And Pressure<br />
			Preset Programs - 4</td>
		</tr>
		<tr>
			<td>PRESSURE RANGE AND SETTINGS</td>
			<td>Delivered Pressure (cm H2O) - 2-50 (Leak Circuit), 0-50 (Valve Circuit)<br />
			Delivered Tidal Volume (cm H2O) - Adult 100 to 2500 ml, Paediatric 50 to 300 ml<br />
			Respiratory Rate - Adult Off 2 to 50 bpm, Paediatric Off 5 to 80 bpm</td>
		</tr>
		<tr>
			<td>DATA</td>
			<td>SpO2 Monitoring - Optional<br />
			FiO2 Monitoring<br />
			7 Days Of High-Resolution Data With 25 Hz Pressure Flow - On Screen And Download<br />
			365 Days Of Summary Data - On Screen And Download</td>
		</tr>
		<tr>
			<td>ALARMS</td>
			<td>A Wide Range Of Settable Alarms Are Available Including FiO2 And SpO2 When Connected</td>
		</tr>
		<tr>
			<td>DIMENSIONS</td>
			<td>Weight - 3.2 Kg<br />
			Dimensions (L x W x H) - 285mm x 215mm x 93mm<br />
			Internal Battery - Lithium-Ion Battery, 14.4V, 6.6 Ah, 95 Wh. 8 Hrs. Backup Power<br />
			Sound Power Level - 43 +/- 3 dBA*<br />
			Sound Pressure Level - 35 +/- 3 dBA*</td>
		</tr>
		<tr>
			<td>IEC 60601-1 CLASSIFICATION</td>
			<td>Medical Electrical Equipment - General Requirements For Basic Safety And Essential Performance -<br />
			Class II (Double Insulation) Type BF Continuous Operation Suitable For Use With Oxygen</td>
		</tr>
		<tr>
			<td>Available</td>
			<td>Rental &amp; Sale</td>
		</tr>
	</tbody>
</table>
